This entry was probably created when you connected the cDAQ without the proper NI drivers being installed. Message 6 of It shows you how to take voltage readings, set up timing and triggering, log to file, and display it with a proper stop not an Abort. Message 5 of 7. I was able to load just DAQmx on a different computer. The simulation will tend to run for a maximum of ms at best.
|Date Added:||9 May 2010|
|File Size:||18.90 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I agree show some code.
Note, this only uses software timing, recently 10Hz. To find the hardware IDs, right click on the device and select properties, then go to the Details tab and select Hardware ID. Thanks again for the robot demo the other night. This is ni cdaq-9174 an issue at low rate, but still not nice. Auto-suggest helps you ni cdaq-9174 narrow down your search results by suggesting possible matches as you type. Most Active Software Boards: This entry was probably ni cdaq-9174 when you connected the cDAQ without the proper NI drivers being installed.
I have uninstalled and reinstalled all the national instruments software with no change. Most Active Hardware Boards: Message 3 of 7. I will have a look at the ni cdaq-9174 you’ve discussed.
Message 2 of 7. It does this even if asked to record 1k’s worth of samples at a given frequency.
Message ni cdaq-9174 of 7. Message 4 of Also, if you want to do more, you should really learn LabView basics first.
It shows you how to ni cdaq-9174 voltage readings, set up timing and triggering, log to file, and display it with a proper stop not an Abort. Ni cdaq-9174 am not sure about this however, maybe someone with better knowledge in DAQmx could confirm this?
NI cDAQ | MySolidWorks 3D CAD Models
Message 2 of We did not end up reinstalling Ni cdaq-9174, so I tried what you suggested. Also, I think it is always advisable to ni cdaq-9174 the “Start Task” and “Stop Task” VIs before and after the while loop, because otherwise cdaq-9714 task is started and stopped repeatedly inside the while loop.
Message 3 of Message 7 of If you have ni cdaq-9174, you get free cdaq-914 to the self-paced online learning material Core ni cdaq-9174, etc You will not be able to configure the task to read both of your channels in one task because they are on two cards that aren’t syncronized, and besides you wouldn’t want to.
Message 5 of 7.
Message 1 of 7. Auto-suggest helps you quickly narrow down your cfaq-9174 results by suggesting possible matches as you type. I have ni cdaq-9174 repairing all of the NI Daqmx 9. Hopefully, ni cdaq-9174 will be a less painful way of addressing the issue. And you can try the Voltage – Continuous Input.
Temperature will come in much slower than in normal AI. Follow the steps in this KB completely deleting the Data folder in step 7: Hello, I’m not sure ni cdaq-9174 this is the right board, but here goes. William Schoettler is working on this SR: